HAPC Does Not Detect Pump / Module
Pump Module Does Not Show up on HAPC
The HAPC requires that the pump address of any Pump / Satellite Module / OEM pump connected to it be set to Pump Chain Address 1, but for use with a computer, the Pump Chain Address needs to be set for 0.
Our Satellite Modules are designed for use with the HAPC, and ship with the address set to 1 by default, but all of our other Syringe Pumps are intended to be capable of stand alone operation, and set to Pump Address of 0.
Therefore, to use these with an HAPC it is necessary to change the Pump Chain Address.
Changing the Pump Chain Address
To change the Pump Chain Address, power up the pump and connect it to a PC using wither USB or RS232.
then, using a terminal emulator such as Pump Terminal, connect to the pump and send the command:
Now power cycle the HAPC. when it turns on, it should detect the pump.
Please sign in to leave a comment.